We occasionally receive emails from customers who feel their balun is defective because they are unable to replicate our test results. This is due in large part to the effect of stray capacitance picked up by the testing method they use. Below you will see a pictorial review of how our testing is performed and the methods used to eliminate the effect of stray capacitance. We also show a scan using a non inductive 50 ohm resistor with wire leads and the magnitude of error this typical method introduces in the results.
Test Setup using High Precision 50 ohm Load
The scan in Figure 1 shows the results using an AIM 4170UHF network analyzer utilizing a high precision encapsulated/shielded 50 Ohm load.
Test setup utilizing a 50 ohm non inductive resistor w/ wire leads
This is the same scan with the only difference being the 50 ohm load now has wire leads. Note the dramatic impact of stray capacitance picked up by the leads of the resistor. The effect is even greater if the model being tested has studs on the sides or top which increases the capacitance pick up area and resulting impedance mismatch.
For those interested, here is the full layout of our test setup using the AIM 4170. We should also note the impact of this capacitance is the most noticeable on 1:1 baluns. The complex winding of other ratios will reduce or eliminate the effect of this capacitance on test results.